184.108.40.206 Accessory equipment.
used in conjunction with measurement receivers shall not degrade measurement
220.127.116.11 Excess personnel and equipment.
area shall be kept free of unnecessary personnel, equipment, cable racks,
and desks. Only the equipment essential to the test being performed shall be
in the test area or enclosure. Only personnel actively involved in the test
shall be permitted in the enclosure.
18.104.22.168 Overload precautions.
receivers and transducers are subject to overload, especially receivers
without preselectors and active transducers. Periodic checks shall be
performed to assure that an overload condition does not exist.
Instrumentation changes shall be implemented to correct any overload
22.214.171.124 RF hazards.
Some tests in this standard
will result in electromagnetic fields which are potentially dangerous to
personnel. The permissible exposure levels in DoDI 6055.11 shall not be
exceeded in areas where personnel are present. Safety procedures and devices
shall be used to prevent accidental exposure of personnel to RF
126.96.36.199 Shock hazard.
Some of the tests require
potentially hazardous voltages to be present. Extreme caution must be taken
by all personnel to assure that all safety precautions are observed.
188.8.131.52 Federal Communications Commission (FCC)
Some of the tests require high level
signals to be generated that could interfere with normal FCC approved
frequency assignments. All such testing should be conducted in a shielded
enclosure. Some open site testing may be feasible if prior FCC coordination